In order to accurately measure the photon flux and to assist in aligning the beam, we have designed a modified beamstop device based on a photo diode integrated with the beamstop. The beamstop contains a small CdWO4 crystal that completely stops the X-rays and at the same time produces photoluminescence proportional to the X-ray flux. The light is then guided to a photosensitive diode using a flexible light pipe to monitor the flux. With this device we achieve the goal of stopping the primary X-ray beam and simultaneously monitoring the X-ray intensity, thus eliminating the need for integrating ion-chambers into the capillary or collimator mount.
|Original language||English (US)|
|Number of pages||3|
|Journal||Nuclear Instruments and Methods in Physics Research, Section A: Accelerators, Spectrometers, Detectors and Associated Equipment|
|State||Published - Sep 1 2011|
- Synchrotron radiation
ASJC Scopus subject areas
- Nuclear and High Energy Physics